Titaniseret Mesh Hvilke slags mesh? - og til hvilke typer operationer? Forskellige vægtklasser Mesh Mesh ekstra light = 16 g. Mesh light = 35 g. Mesh strong = 65 g.
Titaniseret Mesh Klinisk dokumenteret effekt Sheidbach et al. (Europ. Surg. Res. 2004): “Influence of titanium coating of a heavy weight polypropylene mesh” “(...) Since the less pronounced inflammatory reaktion and resulting more favorable shrinkage properties seen in this study were achieved with an identical amounth of implanted material, it must be concluded that the effect is due entirely to the titanium coating/surface structure.” Neymeyer et al. (2006): Animal experiment-based investigation of the compatibility of titanium-coated polypropylene mesh implants for urogynecological plasties under conditions of accompanying intestine surgery” “(...) The connective tissue embedding that occurred as a result of the foreign body reaction was significantly less in the case of titanium-coated meshes that non-titanium-coated implants... Sheidbach et al. (Europ. Surg. Endosc. 2004): “Influence of titanium coating of a heavy weight polypropylene mesh” “(...) Reports in the literature show that heavyweight polypropylene meshes undergo shrinkage - as a result of inflammatory reactions and scar formation - of up to 20-50% of their original size [10,15]. In comparison, the shrinkage of the Atrium mesh in our experiment was, at 15%, somewhat more favorable. In the case of the titanium-coated mesh, however, the shrinkage characteristics were even more favorable (8.8%)...” Se evt. også studier af: Sheidbach et al. (Surg. Endosc. 2003) Schug - Pass et al. (Surg. Endosc 2006) Horstmann et al. (World Surgery Journal 2006) Kock et al. (British Journal of Surgery 2008)