GL900 USER'S MANUAL - Graphtec
Settings and Measurement 3.4 Setting Menus When you press the MENU key during Free Running, the following menu screens appear. The menu screens are classified by the tab for each setting item. (1) AMP settings This menu is used to specify input signal-related settings. Setting Selections available Input Off, Voltage, Temperature, Humidity Range [Voltage] 20, 50, 100, 200, 500 mV; 1, 2, 5, 10, 20,50,100,200,500V,1-5V [Temperature] TC-K, TC-J, TC-T, TC-R, TC-E, TC-B, TC-S, TC-N, TC-W Filter Off, Line, 5, 50, 500Hz EU Function Off, On (Scaling Measured Upper limit value Numeric value input settings) value Lower limit value Numeric value input EU output Upper limit value Numeric value input value Lower limit value Numeric value input Decimal point 1, 10, 100, 1000, 10000 Unit selection Current, length, area, volume, speed, acceleration, frequency, weight, power, pressure, flow, temperature Unit (Selections vary depending on the selected units listed in the above.) User-defined unitText input Misc. Span Upper limit value Numeric value input setting Lower limit value Numeric value input Annotation setting Text input (31 characters max.) Waveform color setting Red/Green/Blue (RGB) 0 to 31 available for each color Perform Auto Zero ADJ. Execute Reset Auto Zero ADJ. Execute Logic and Pulse Off, Logic, Pulse [Logic] Off, On [Pulse] Input Off, Revolution counts, Counts, Inst. Filter Off, On Slope ↑ H, ↓ L EU Function Off, On Measured value Numeric value input EU output value Numeric value input Unit selection Current, length, area, volume, speed, acceleration, frequency, weight, power, pressure, flow, temperature Unit (Selections vary depending on the selected units listed in the above.) User-defined unit Text input 3-17
Settings and Measurement Switching displays Analog and Logic/Pulse can be switched as shown below. • Display Logic/Pulse Data A screen switches with ENTER key. • Display Analog Data Analog settings This screen allows you to set conditions for analog signals. CHECKPOINT When selecting "All Ch" at the Input and the Filter, all channels are set to the same setting. For the Range, the channels which are set to same setting of the Input are set to same range setting. For the all channels Span setting, the channels which are set to same range setting are set to same span setting. (1)-1 Input Selects input conditions. Selection Off Voltage Temperature Humidity Description Disables the measurement of input signals and the waveform and digital display. Used for measuring direct-current voltages. Used for measuring temperatures. Used for measuring humidity with humidity sensor B-530. This selection sets the voltage range to 1V and disables EU settings. (1)-2 Range Specifies the range of signal input to be measured. Input setting Selection Voltage 20, 50, 100, 200, 500mV; 1, 2, 5, 10, 20, 50, 100, 200, 500V; 1-5V Temperature TC-K, TC-J, TC-T, TC-R, TC-E, TC-B, TC-S, TC-N, TC-W, Humidity No selection 3-18
Index Level Operations ............
503-10 Shinano-cho, Totsuka-ku, Yok