Using the AC Adapter • The Analyzer can be operated on 110 V AC power using the power adapter. • The power supply and battery shell are permanently connected via the adapter cord. • The battery shell fits into the IRMA TRUpoint battery compartment. • The power cord connects the power supply to the electrical wall outlet.
Cartridge Equilibration • Remove cartridges from shipping container and store as follows prior to use: Cartridge(s) Equilibration Time Storage Temperature Warm up Time BG, CC 72 hours 15-30°C (59-86°F) None H3, H4, GL, LA 1 hour 15-30°C (59-86°F) None CR None 2-8°C (35.6-46.4°F) 15 min • If storage range fluctuates more than 8°C (14.4°F), cartridges must go through an additional equilibration prior to use. • Cartridges can be used in temperatures down to 12°C, 54°F if used within 4 hours of transfer from the 15-30°C storage.