Theory of Operation www.ti.com The behavior of resistance for PT100 RTD from –200°C to 850°C is shown in Figure 28. 400 360 320 RTD( Temp) R LINFIT ( Temp) Resistance (Ohms) 280 240 200 160 120 80 40 0 200 95 10 115 220 325 430 535 640 745 850 Temperature (C) Figure 28. RTD Resistance and Linear Fit Line versus Temperature The change in RTD resistance is piecewise linear over small temperature ranges. An end-point curve fitting shows RTDs have a second-order, nonlinearity of approximately 0.375% as shown in Figure 29 . To correct this correct this nonlinearity error requires implementing a digital linearization technique. 20 Resistance Error and Nonlinearity %FSR 5 18 16 4.5 4 Resistance Error (Ohms) 14 3.5 12 3 Nonlin Ohms ( Temp) 10 2.5 Nonlin Error ( Temp) 8 2 6 1.5 Resistance Nonlinearity (%) 4 2 1 0.5 0 200 95 10 115 220 325 430 535 640 745 0 850 Temp Temperature (C) Figure 29. Resistance Error and Nonlinearity versus Temperature 26 Temperature Sensor Interface Module for Programmable Logic Controllers TIDU271–May 2014 (PLC) Submit Documentation Feedback Copyright © 2014, Texas Instruments Incorporated

www.ti.com Theory of Operation The three techniques for RTD linearization are as follows: 1. Simple Linear Approximation Method 2. Piecewise Linear Approximation Method 3. Callendar-Van Dusen Equations Method A brief comparison is also given in Table 6. Table 6. Linear Approximation Methods METHOD ADVANTAGE DISADVANTAGE OTHER REMARKS Simple Linear Approximation Piecewise Linear Approximation Callendar-Van Dusen Equations Easy to implement Fast execution Accurate over small temperature ranges Smaller code size Accuracy depends on the entries in the lookup table Fast execution Most accurate Least accurate over wide temperature range High accuracy More entries in lookup table Bigger code size Extremely Processor Intensive Time consuming Requires Square root and Power functions on the processor Simple method when memory size is limited and temperature range is smaller Most practical method implement by most of the embedded systems Not practical in most applications This reference design implements the piecewise linear approximation method using the manufacturer’s lookup table with more number of entries to achieve better accuracy. 5.2.2 RTD Resistance Measurement The basic principle required for RTD measurement is based on Ohm’s Law. Ohm's law equation is as follows: V R I where • R = Resistance of the RTD element • I = Known excitation current • V = Voltage across RTD element (16) As an RTD is a passive sensor, an RTD does not produce output on its own. An RTD requires a known constant current source for the RTD's'excitation to produce a useful voltage output proportional to the resistance of the RTD. The resulting voltage output is then amplified and fed to an ADC for measurement. By this method, an unknown resistance of an RTD can be easily calculated and then transformed to temperature using a software algorithm. Depending on the nominal resistance of the RTD, different excitation currents can be used. The excitation current flow through an RTD should be minimized to avoid self-heating. In general, approximately 1 mA or less current is used. I1 V IN+ R RTD V DIFF GAIN ΔΣ ADC V IN- Figure 30. Simplified RTD Measurement Circuit TIDU271–May 2014 Submit Documentation Feedback Temperature Sensor Interface Module for Programmable Logic Controllers (PLC) Copyright © 2014, Texas Instruments Incorporated 27

- Page 1 and 2: TI Designs Temperature Sensor Inter
- Page 3 and 4: www.ti.com 2 Overview Overview The
- Page 5 and 6: www.ti.com 5 Theory of Operation Th
- Page 7 and 8: www.ti.com Theory of Operation 5.1.
- Page 9 and 10: www.ti.com Theory of Operation Each
- Page 11 and 12: www.ti.com Theory of Operation The
- Page 13 and 14: www.ti.com Theory of Operation Figu
- Page 15 and 16: www.ti.com Theory of Operation Use
- Page 17 and 18: www.ti.com Theory of Operation 5.1.
- Page 19 and 20: www.ti.com F F CM _ 1 CM _ 2 1 2
- Page 21 and 22: www.ti.com Theory of Operation 5.1.
- Page 23 and 24: www.ti.com 5.2 RTD Input Channel Th
- Page 25: www.ti.com Table 5. RTD Class and T
- Page 29 and 30: www.ti.com Theory of Operation 5.2.
- Page 31 and 32: www.ti.com Theory of Operation The
- Page 33 and 34: www.ti.com Theory of Operation 5.2.
- Page 35 and 36: www.ti.com Theory of Operation Figu
- Page 37 and 38: www.ti.com Theory of Operation I1 R
- Page 39 and 40: www.ti.com Theory of Operation 5.2.
- Page 41 and 42: www.ti.com Protection Use the TVS r
- Page 43 and 44: www.ti.com NOTE: Power Supply Desig
- Page 45 and 46: www.ti.com Power Supply Design and
- Page 47 and 48: www.ti.com Test Setup, Calibration,
- Page 49 and 50: www.ti.com Test Setup, Calibration,
- Page 51 and 52: www.ti.com Test Setup, Calibration,
- Page 53 and 54: www.ti.com Bill of Materials Table
- Page 55 and 56: www.ti.com Bill of Materials Table
- Page 58: 1 1 TP5 1 1 TP7 300 R20 MISO_ISO SC
- Page 62: 1 2 2 2 1 1 1 2 2 2 2 2 1 1 1 1 1 2
- Page 66 and 67: www.ti.com 11 PCB Layer Plots To do
- Page 68 and 69: www.ti.com PCB Layer Plots Figure 8
- Page 70: IMPORTANT NOTICE FOR TI REFERENCE D

Nothing to read

Loading...

Thank you, for helping us keep this platform clean.

The editors will have a look at it as soon as possible.

Loading...

Loading...

Are you sure you want to delete your template?

**Magazine:** tidu271

Start using Yumpu now!